- Case Report
- Open Access
Palatal erosion and oronasal fistulation following covered nasopharyngeal stent placement in two dogs
© Cook et al.; licensee BioMed Central Ltd. 2013
- Received: 1 October 2012
- Accepted: 17 April 2013
- Published: 1 May 2013
Treatment options for dogs with nasopharyngeal stenosis include fluoroscopic placement of metallic stents. Reported complications include entrapment of hair and food, obstruction and persistent nasal discharge. Two toy breed dogs were examined for persistent nasal discharge and halitosis at 4 and 20 months after placement of permanent metallic stents for acquired nasopharyngeal stenosis. Full thickness defects were found in the palate of both dogs, with extensive communication between the mouth and the nasal passages. Portions of the metal stent were observed within the lesion in both patients. Additional treatment was declined by the owner of one dog; the stent was removed through the fistula in the other dog. Palatal erosion with secondary oronasal fistulation is a potential complication of nasopharyngeal stent placement in dogs.
- Nasopharyngeal stenosis
Nasopharyngeal stenosis secondary to regurgitation of gastric contents during anesthesia has rarely been reported in dogs [1, 2]. Clinical signs associated with stenosis of the nasopharynx can be severe and the quality of life for affected animals is substantially compromised. Historically, attempts to alleviate this condition with surgery, endoscopic ballooning or laser ablation have been variably successful, and may in fact trigger progression from partial stenosis to complete occlusion [1, 3–5].
In recent years, excellent short and long term outcomes have been reported with the use of metal stents for both dogs and cats with nasopharyngeal stenosis . Patients with complete obstruction of the affected area appear predisposed to ingrowth through an open stent, so covered stents are recommended for these individuals.
Complications associated with nasopharyngeal stents have been reported previously and include exaggerated swallowing, partial stent obstruction with fur, stent obstruction with tissue, and persistent nasal discharge . The dogs described in this report developed full thickness palatal defects several months following nasopharyngeal stent placement. To the authors’ knowledge, this complication has not been previously reported.
A 3-year-old spayed female Chihuahua weighing 3.4-kg (7.5-lb) was examined at the Veterinary Medical Teaching Hospital (VMTH) at Texas A&M University in February 2012 for sneezing and nasal discharge, described as brown and mucoid. Clinical signs began approximately 4 months prior to presentation and were variably responsive to antimicrobial therapy prescribed by the referring veterinarian, including cefovicin, enrofloxacin, amoxicillin-clavulanic acid, and sulfadimethoxine-ormetoprim. Diphenhydramine and prednisone were also administered without clinical improvement. In the two weeks prior to this visit, the discharge had increased substantially and the dog’s appetite had decreased.
When examined in February 2012, the dog was quiet and reserved, with a body condition score of 4/9. Rectal temperature was 100.9°F, pulse was 140 beats/min and respiratory rate was 24 breaths/min. A grade II/VI left-sided systolic murmur was noted; there were no arrhythmias or pulse deficits. The remainder of the physical examination within normal limits. There was a strong odor noted, emanating from the dog’s nasal area and mouth. Purulent discharge was noted around both nares. Airflow was assessed using a glass slide and appeared to be decreased through both nasal passages. The dog would not permit examination of the oral cavity.
Results of a complete blood count and serum electrolyte panel were within normal limits. A three view thoracic radiographic study was within normal limits. A small volume of fluid was noted in the caudal thoracic esophagus.
Aerobic culture of the exudate was performed, and a mixed bacterial population was cultured, including an extended spectrum beta lactamase positive Escherichia coli, Pseudomonas aeruginosa and an alpha hemolytic Streptococcus species. Surgical options, including removal of the stent and closure of the palate were discussed with the owner, but declined. The dog was euthanized 8 weeks later.
A 2-year-old spayed female Chihuahua weighing 2.8-kg (6.2-lb) was examined at the Small Animal Hospital at University of Florida in July 2011 for persistent nasal discharge and open mouth breathing. The patient underwent routine ovariohysterectomy in June 2010; regurgitation of gastric contents was noted during the procedure. Bilateral nasal discharge and stertorous breathing began soon after surgery. The dog was initially treated with a variety of antibiotics with variable response. CT performed in November 2010 revealed nasopharyngeal stenosis, with concurrent rhinitis and turbinate lysis. A dorsal rhinotomy with complete turbinectomy was performed in December 2010. In February 2011, the dog was referred to University of Florida. CT, rhinoscopy and positive contrast rhinography were performed and confirmed nasopharyngeal obstruction. An osseous component to the nasopharyngeal stenosis was present with only 1.1 mm of distance between the palatine bone and the sphenoid bone. The nasal septum, rostral and mid-cavity turbinates were absent, consistent with the previous surgery. Because of the bony abnormality associated with the nasopharyngeal stenosis, a ventral rhinotomy was elected. The dog underwent general anesthesia and pharyngeal intubation. The rhinotomy was performed through a midline incision. The incision extended through the caudal hard palate and into the rostral soft palate, leaving the caudal aspect of the soft palate intact. A pneumatic burr was used to remove a window in the caudal aspect of the palatine bone. Purulent material was present in the choana and was removed. Soft tissue was removed with a combination of sharp and blunt dissection. The pneumatic burr was used to create a trough through the sphenoid bone. The trough was gradually enlarged until an 18 French red rubber catheter could be placed through the bony trough and choana. The red rubber catheter was advanced to extend through the stenotic area to the right nostril, and was sutured flush to the skin to serve as a temporary nasopharyngeal stent. The nasal mucosa was closed with 5–0 polydioxanone in a simple continuous pattern and the oral mucoperiosteum was closed with simple continuous pattern, using 5–0 polydioxanone. Immediate postoperative therapy included methadone (0.2 mg/kg [0.9 mg/lb] IV every 6 hours) and cefazolin (22 mg/kg [10 mg/lb] IV every 8 hours). The dog was discharged from the hospital with tramadol (2.1 mg/kg [1 mg/lb] PO every 8 hours for 5 days), and cephalexin (30 mg/kg [13.6 mg/lb] PO every 12 hours for 14 days).
A 8 mm × 30 mm covered self-expanding metallic stenta was ordered. Stent diameter was determined from the size of the red rubber catheter (i.e. 6 mm) plus an additional 2 mm as previously described . The stent diameter was determined from the red rubber catheter because the sphenoid bone was modified during surgery and the CT was performed prior to surgery. Stent length was calculated from CT images. The dog returned 6 weeks later and was placed under general anesthesia for stent placement. A guidewire was inserted retrograde through the red rubber catheter. The sutures were removed from the red rubber catheter and it was removed. A balloon catheter was advanced over the guidewire, into the nasopharynx and inflated to 8 mm wide. The balloon was removed and the stent was positioned and deployed using fluoroscopic guidance. The dog improved immediately postoperatively. The dog was administered prednisone (0.44 mg/kg [0.2 mg/lb] PO daily for three days then every other day for 5 days), cephalexin (30 mg/kg [13.6 mg/lb] PO twice daily for 14 days), and colchicine (0.035 mg/kg [0.016 mg/lb] PO daily for 14 days). Colchicine was administered to in an attempt to prevent the formation of granulation tissue at the ends of the stent. The dog was discharged from the hospital with a recommendation to perform nebulization for 10 minutes duration 3–4 times daily with a commercial nebulizer.
Nearly one year after removal of the stent, the dog is still able to breathe through the nose. Intermittent mucopurulent nasal discharge and nasal congestion are reported. The owners must hand feed the dog to prevent impaction of food in the oronasal fistula, in part due to the dog’s rapid eating style.
The authors are not aware of any prior reports describing this complication to stent placement in dogs with acquired nasopharyngeal stenosis. The reason for erosion through the palate is unclear and a direct causal relationship to the metal stent can only be inferred. It is likely however that the stent played a role in development of the palatal defect.
In the first dog described, a guarded needle was used to establish an opening through the stenotic area prior to stent placement. The device was guided fluoroscopically, and appeared appropriately positioned. However, the needle may have been driven through the tissue slightly more ventrally than intended or at an angle, thereby damaging the palate. In the second dog reported here, the palate may have been compromised by the previous surgery, ultimately resulting in pressure necrosis through the tissue. In this dog, it was deemed necessary to perform a ventral rhinotomy to modify the bony abnormalities and provide space for the permanent stent. The ventral rhinotomy procedure and permanent stent placement were staged by six weeks but that interval may not have been long enough for complete healing and establishment of healthy tissue. The two procedures were staged to allow healing of the palate prior to placement of the permanent stent, as the permanent stent would place outward radial force onto the incision site possibly causing dehiscence.
Both covered and uncovered nasopharyngeal stents have been described in veterinary patients. Covered stents are recommended in patients with complete stenosis, as closure has been reported following the use of open stents in this patient population . Covered stents were placed in both dogs in this report, and it is possible that this choice contributed to palatal erosion. Underlying tissue is probably more vulnerable to infection or other compromise underneath a covered stent and therefore more likely to breakdown. A self-expanding stent was placed in the second dog reported here. This was chosen rather than a balloon-expanding stent because a balloon-expanding, covered stent was not available in the size required.
Stent fracture has been reported in dogs with nitinol tracheal stents, and is associated with inflammation of the adjacent tissues . The aboral end of the stent in dog 2 appears to have fractured, and probably contributed to the lesions on the tongue. It is unclear if the stent fractured prior to palatal erosion, or subsequent to that event. Stent fracture would certainly increase the risk of damage to adjacent tissue and may have contributed to the palatine defect. The causes for stent fracture or fraying of the ends of a stent are unclear, but excessive movement or deformity may play a role.
It is interesting to note that both patients reported here were small Chihuahuas. Because of technical limitations, stents may be somewhat oversized in toy breeds if a covered device is required. In small dogs, it is more likely for a substantial portion of the stent to extend into the nasopharynx overlying the soft palate. This tissue is probably subject to substantial compressive forces during swallowing, which may contribute to structural failure of the stent.
Additionally, the skull of a Chihuahua has a unique shape. In dog 2, bony abnormalities were evident on CT examination and at surgery. Such bony restrictions may force the stent to expand only in the ventral direction placing excessive pressure on the palate. There may also be increased mobility of the distal end if the stent is moderately pinched in the stenotic region and therefore unable to expand appropriately.
Erosion of tissue has been reported with stent placement in the respiratory and gastrointestinal tracts in human patients and may be related to stent integration into surrounding tissues [7–9]. Recommendations typically include stent removal if possible, antibiotic therapy and surgical repair of the damaged tissue.
Dog 2 experienced improvement of clinical signs despite the complications associated with the nasopharyngeal stent. The oronasal fistula that was formed by the stent was not addressed surgically and did not close with time despite stent removal. The fistula allowed drainage of nasal mucous into the mouth. Additionally, the dog was able to move air through the nares and bypass the stenotic area. No further rhinogram or rhinoscopy were performed, so it is unknown if the dog moves air thorough the choana, the ornasal fistula or both.
Although nasopharyngeal stenting using permanent expandable metal stents may be an effective solution for patients with nasopharyngeal stenosis, palatal erosion with oronasal fistulation is a possible complication and should be included in any discussion about stent placement.
aInfiniti Medical LLC, Menlo Park, CA.
bOlympus America Inc, Center Valley, PA.
cOlympus America Inc, Center Valley, PA.
Dr. Mankin’s present address is Department of Small Animal Clinical Sciences, College of Veterinary Medicine and Biomedical Sciences, Texas A&M University, College Station, TX 77843.
- Berent AC, Weisse C, Todd K, Rondeau MP, Reiter AM: Use of a balloon-expandable metallic stent for treatment of nasopharyngeal stenosis in dogs and cats: six cases (2005–2007). J Am Vet Med Assoc. 2008, 233: 1432-1440. 10.2460/javma.233.9.1432.View ArticlePubMedGoogle Scholar
- Berent AC, Kinns J, Weisse C: Balloon dilation of nasopharyngeal stenosis in a dog. J Am Vet Med Assoc. 2006, 229: 385-388. 10.2460/javma.229.3.385.View ArticlePubMedGoogle Scholar
- Coolman BC, Marretta SM, McKiernan BC, Zachary JF: Choanal atresia and secondary nasopharyngeal stenosis in a dog. J Am Anim Hosp Assoc. 1998, 34: 497-501.View ArticlePubMedGoogle Scholar
- Hunt GB, Perkins MC, Foster SF, Barrs VR, Swinney GR, Malik R: Nasopharyngeal disorders of dogs and cats: a review and retrospective study. Comp Cont Educ Pract Vet. 2002, 24: 184-199.Google Scholar
- Kirbertgre RM, Steenkamp G, Spotswood TC, Boy SC, Miller DB, van Zyl M: Stenotic nasopharyngeal dysgenesis in the Dachshund; seven cases (2002–2004). J Am Anim Hosp Assoc. 2006, 42: 290-297.View ArticleGoogle Scholar
- Sira PA, Krahwinkel DJ: Self-expanding nitinol stents for the treatment of tracheal collapse in dogs: 12 cases (2001–2004). J Am Vet Med Assoc. 2008, 232: 228-263. 10.2460/javma.232.2.228.View ArticleGoogle Scholar
- Saafan ME: Endoscopic management of congenital bilateral posterior choanal atresia: value of using stents. Eur Arch Otorhinolaryngol. 2013, 270: 129-134. 10.1007/s00405-012-1956-0.View ArticlePubMedGoogle Scholar
- Lund ME: Airway stenting for patients with benign airway disease and the Food and Drug Administration advisory. Chest. 2007, 132: 1107-1108. 10.1378/chest.07-0242.View ArticlePubMedGoogle Scholar
- Wells WJ, Hussain NS, Wood JC: Stenting of the mainstem bronchus in children: a word of caution. Ann Thor Surg. 2004, 77: 1420-1422. 10.1016/S0003-4975(03)00893-2.http://dx.doi.org/10.1007/s00405-012-1956-0,View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.